URGENT MEDICAL DEVICE RECALL
Aero Healthcare, after consultation with the Therapeutic Goods Administration (TGA), is conducting a Medical Device Recall of the above PE Burn Sheets. We are contacting you as the potentially affected product may have been supplied to your organisation.
|TGA Reference Number||:||RC – 2022 – RN – 00507 – 1|
|Product||:||AEROBURN PE Burn Sheets 60 cm x 90 cm|
|Batch Numbers||:||186090, 196090, BS690323, BS690916, BS690D14, BS690N08, BS690O10, BS691302, BS691321, BS691625, BS691720 and BS691810 (supplied individually or as part of a first-aid kit, see Appendix 1).|
|Intended Use||:||A sterile, trauma covering used to maintain the stability of a burn patient en-route to a burn centre or hospital. The device is intended to be applied at the emergency site and is used to prevent burn progression, minimise trauma, and provide protection against contamination.|
ABS6090 is a sterile device. The product is packed in a sealed pouch providing a microbiological barrier to maintain product sterility over its shelf-life. During handling, if any abnormal forces are applied to a specific sealed edge, the pouch may open resulting in loss of microbiological barrier and sterility, potentially leading to a secondary infection.
This problem does not affect any batches of ABS6090 that were supplied prior to September 2018. The product is supplied individually or as part of a first aid kit (see full list of affected kits in Appendix 1).
- If you are a user, please inspect the pouch carefully (particularly at the edge opposite the ‘peel to open’ edge) and do not use the product if the pouch sealing is broken as shown in images below.
- If you have to discard any product, call Aero Healthcare on 1800 628 881 to organise a replacement.
- If you are selling or distributing this product, ensure your relevant staff are aware of this recall notice. Please advise your customers of this problem and provide them with a copy of this letter. (Download PDF Version of ABS6090 Customer Letter.pdf )
We apologise for any inconvenience this notification may cause and if you have further questions, please do not hesitate to contact Aero Healthcare on 1800 628 881 or email@example.com.
Appendix 1: List of first aid kits containing ABS6090
First aid kit reference code